Non-clinical testing has demonstrated that the UroLift Implant is MR Conditional. A patient with this device can be safely
scanned in an MR system immediately after placement meeting the following conditions1:
Under the scan conditions defined above, the UroLift Implant is expected to produce a maximum temperature rise of 2.4°C after 15 minutes of continuous scanning (i.e., per pulse sequence)
In non-clinical testing, the image artifact caused by the device extends approximately 15 mm from the UroLift Implant when imaged with a gradient echo pulse sequence and a 3.0 Tesla MRI system.1
The safety of the delivery system has not been evaluated in the MR environment, and therefore, the delivery system should
not be used within the MR environment.